This item is no longer available. Suggested replacements. UVAS sc Probe, 1 mm, with SC Controller · Get Quote. Continuous UV Results 1 – 20 of Search results for sc Amtax sc Ammonium Analyzer, mg/L NH₄-N, V, 5 m Filterprobe sc. This item is no longer available. Suggested replacements. UVAS sc Sensor, 5 mm, with SC Controller · Get Quote. Continuously protect plant treatment.
|Published (Last):||28 August 2007|
|PDF File Size:||16.95 Mb|
|ePub File Size:||5.99 Mb|
|Price:||Free* [*Free Regsitration Required]|
Wear cotton gloves to protect your hands and to avoid fingerprints on the bulb. The passcode is factory set to sc the five digits must be followed by a space to remove the trailing asterisk.
Controller Dimension Illustrations Installation 3. Page 71 Index Sample Line Installation Page 23 Allow at least mm Section 10 Repair Service Authorization must be obtained from Hach Company before sending any items for repair. Pay attention to all danger and caution statements. The data logs store the measurement data at selected intervals.
Hach-Lange SC 100 User Manual
Problem resolution and fuse replacement should only be performed by qualified service personnel. The incident light Method of Detection beam is composed of a tungsten filament light source with a color temperature between and K. Depends on sample Clean the sensor characteristics.
Page 20 Installation 3. The calibration history log shows the gain value, the time and date of the calibration, and the initials of the operator performing verification. Tug gently to check the connection.
After all sc010 and electrical connections have been completed and checked, place the head on the body and supply power to the system. Table of Contents Add to my manuals Add.
Under normal use, Hach recommends replacing the lamp once a year to maintain peak performance. The display will flash on startup, when the hold outputs function has been activated, and when the filter function signal average is changed to a different value.
HACH SC E INSTRUMENT MANUAL Pdf Download.
Troubleshooting Section 7 Hsch Calibrate Sensor as required by regulatory agency Per agency-dictated schedule. Do not touch the new bulb with bare hands. Disconnect and remove the existing wires between the quick connect and terminal block J5, see Figure 21 on page 4. Section 7 Troubleshooting Section 7 Troubleshooting 7.
Relay loads must be resistive. Failure to do so could result in serious injury to the operator or damage to the equipment. Troubleshooting Table 11 presents sensor warnings displayed in the Event Log, possible causes, and corrective actions.
Rinse with deionized water sc1000 calibrating. By pressing ‘print’ button you will print only current page. Troubleshooting Troubleshooting Table 11 presents sensor warnings displayed in the Event Log, possible causes, and corrective actions.
Section 5 System Startup 5. Remove the quick connect fitting and wires and install the threaded plug on the opening to maintain the environmental rating. Close and secure the cover.
Hach-Lange SC User Manual | 58 pages
Page 50 Maintenance a. The time is available only in hour military format.
Failed fuses are an indication that an equipment problem could exist.